* This is the Professional Version. *
- Uses of MRI
- Variations of MRI
- Disadvantages of MRI
- Resources In This Article
- Drugs Mentioned In This Article
Magnetic Resonance Imaging
MRI uses magnetic fields and radio waves to produce images of thin slices of tissues (tomographic images). Normally, protons within tissues spin to produce tiny magnetic fields that are randomly aligned. When surrounded by the strong magnetic field of an MRI device, the magnetic axes align along that field. A radiofrequency pulse is then applied, causing the axes of many protons to momentarily align against the field in a high-energy state. After the pulse, protons relax and resume their baseline alignment within the magnetic field of the MRI device. The magnitude and rate of energy release that occurs as the protons resume this alignment (T1 relaxation) and as they wobble (precess) during the process (T2 relaxation) are recorded as spatially localized signal intensities by a coil (antenna) built within the MRI device. Computer algorithms analyze these signals and produce detailed anatomic images.
The relative signal intensity (brightness) of tissues in an MRI image is determined by factors such as
By controlling the radiofrequency pulse and gradient waveforms, computer programs produce specific pulse sequences that determine how an image is obtained (weighted) and how various tissues appear. Images can be
For example, fat appears bright (high signal intensity) on T1-weighted images and relatively dark (low signal intensity) on T2-weighted images; water and fluids appear relatively dark on T1-weighted images and bright on T2-weighted images. T1-weighted images optimally show normal soft-tissue anatomy and fat (eg, to confirm a fat-containing mass). T2-weighted images optimally show fluid and abnormalities (eg, tumors, inflammation, trauma). In practice, T1- and T2-weighted images provide complementary information, so both are important for characterizing abnormalities.
Recently introduced high-resolution MRI scanners increase image quality and diagnostic accuracy.
MRI is preferred to CT when soft-tissue contrast resolution must be highly detailed (eg, to evaluate intracranial or spinal cord abnormalities, inflammation, trauma, suspected musculoskeletal tumors, or internal joint derangement). MRI is also useful for the following:
Vascular imaging: Magnetic resonance angiography (MRA) is used to image arteries with good diagnostic accuracy and is less invasive than conventional angiography. Gadolinium contrast is sometimes used. MRA can be used to image the thoracic and abdominal aorta and arteries of the brain, neck, abdominal organs, kidneys, and lower extremities. Venous imaging (magnetic resonance venography, or MRV) provides the best images of venous abnormalities, including thrombosis and anomalies.
Hepatic and biliary tract abnormalities: Magnetic resonance cholangiopancreatography (MRCP) is particularly valuable as a noninvasive, highly accurate method of imaging the biliary and pancreatic duct systems.
Masses in the female reproductive organs: MRI supplements ultrasonography to further characterize adnexal masses and to stage uterine tumors.
Certain fractures: For example, MRI can provide accurate images of hip fractures in patients with osteopenia.
Bone marrow infiltration and bone metastases
MRI can also be substituted for CT with contrast in patients with a high risk of reactions to iodinated contrast agents.
With MRI, contrast agents are often used to highlight vascular structures and to help characterize inflammation and tumors.
The most commonly used agents are gadolinium derivatives, which have magnetic properties that affect proton relaxation times. MRI of intra-articular structures may include injection of a diluted gadolinium derivative into a joint.
This ultrafast technique (images obtained in > 1 sec) is used for diffusion, perfusion, and functional imaging of the brain and heart. Its potential advantages include showing brain and heart activity and reducing motion artifacts. However, its use is limited because it requires special technical hardware and is more sensitive to various artifacts than conventional MRI.
Functional MRI is used to assess brain activity by location.
In the most common type, the brain is scanned at low resolution very frequently (eg, every 2 to 3 sec). The change in oxygenated Hb can be discerned and used to estimate metabolic activity of different parts of the brain.
Researchers sometimes do functional MRI while subjects do different cognitive tasks (eg, solve a math equation); the metabolically active parts of the brain are presumed to be the structures most involved in that particular task. Correlating brain function and anatomy this way is called brain mapping.
Functional MRI is used primarily in research but is being increasingly used clinically.
Gradient echo is a pulse sequence that can be used for fast imaging of moving blood and CSF (eg, in MRA). Because this technique is fast, it can reduce motion artifacts (eg, blurring) during imaging that requires patients to hold their breath (eg, during imaging of cardiac. pulmonary, and abdominal structures).
MRS combines the information obtained by MRI (mainly based on water and fat content of tissues) with that of nuclear magnetic resonance (NMR). NMR provides information about tissue metabolites and biochemical abnormalities; this information can help differentiate certain types of tumors and other abnormalities.
PET MRI combines functional PET (see Positron Emission Tomography (PET)) with whole-body MRI. T1-weighted and short T1 inversion recovery (STIR) sequences are frequently used. This method is new and is available in only a few major medical centers.
MRI is relatively expensive, requires longer imaging times than CT and may not be immediately available in all areas.
Other disadvantages include problems related to
MRI is relatively contraindicated in patients with implanted materials that can be affected by powerful magnetic fields. These materials include
Ferromagnetic material may be displaced by the strong magnetic field, injuring a nearby organ; for example, displacement of vascular clips can result in hemorrhage. Displacement is more likely if the material has been in place < 6 wk (before scar tissue forms). Ferromagnetic material can also cause imaging artifacts.
Magnetically activated medical devices may malfunction when exposed to magnetic fields.
Magnetic fields may induce current in any conductive materials strong enough to produce enough heat to burn tissues.
Whether a specific device is compatible with MRI depends on the type of device, its components, and its manufacturer (see the MRI safety web site ). Patients with an implantable device should not be placed in the MRI magnetic field until practitioners are sure that MRI is safe with such a device in place. Also, MRI machines with different magnetic field strengths have different effects on materials, so safety in one machine does not ensure safety in another.
The MRI magnetic field is very strong and may always be on. Thus, a ferromagnetic object (eg, an O2 tank, a metal pole) at the entrance of the scanning room may be pulled into the magnet bore at high velocity and injure anyone in its path. The only way to separate the object from the magnet may be to turn off (quench) the magnetic field.
The imaging tube of an MRI machine is a tight, enclosed space that can trigger claustrophobia even in patients without preexisting phobias or anxiety. Also, some obese patients do not fit on the table or within the machine. Premedication with an anxiolytic (eg, alprazolam or lorazepam 1 to 2 mg po) 15 to 30 min before scanning is effective for most anxious patients.
MRI scanners with an open side can be used for patients with claustrophobia (or those who are very obese). Images obtained during open MRI may be inferior to those of enclosed scanners depending on the field strength of the magnet, but they are usually sufficient for making a diagnosis.
Patients should be warned that the MRI machine makes loud, banging noises during scanning.
Gadolinium-based contrast agents injected IV can cause headache, nausea, pain, and distortion of taste, as well as sensation of cold at the injection site.
Serious contrast reactions are rare and much less common than with iodinated contrast agents.
However, nephrogenic systemic fibrosis is a risk in patients with impaired renal function. Nephrogenic systemic fibrosis is a rare but life-threatening disorder that involves fibrosis of the skin, blood vessels, and internal organs, resulting in severe disability or death. For patients with impaired renal function, the risks and benefits of contrast MRI should be weighed; in addition the following is recommended:
Use gadolinium only when necessary and at the lowest dose possible.
Check renal function if diabetes, dehydration, or heart failure is clinically suspected, if patients are taking certain drugs that can cause renal insufficiency, or if patients are elderly. (Patients who have a GFR of < 30 mL/min/1.73 m2 should not be given gadolinium contrast agents. If GFR is between 30 and 60 mL/min/1.73m2, patients can be hydrated IV before contrast administration.
Consider alternative imaging methods.
Drug NameSelect Trade
* This is the Professional Version. *